Chest-abdomen-pelvis CT for staging in cancer patients: dose effectiveness and image quality using automated attenuation-based tube potential selection
© Beeres et al.; licensee BioMed Central Ltd. 2014
Received: 7 March 2014
Accepted: 18 August 2014
Published: 2 September 2014
Evaluation of automated attenuation-based tube potential selection and its impact on image quality and radiation dose in CT (computed tomography) examinations for cancer staging.
A total of 110 (59 men, 51 women) patients underwent chest-abdomen-pelvis CT examinations; 55 using a fixed tube potential of 120 kV/current of 210 Reference mAs (using CareDose4D), and 55 using automated attenuation-based tube potential selection (CAREkV) also using a current of 210 Reference mAs.
This evaluation was performed as a single-centre, observer-blinded retrospective analysis. Image quality was assessed by two readers in consensus. Attenuation, image noise, signal-to-noise ratio (SNR), and contrast-to-noise ratio (CNR) were measured or calculated for objective image evaluation. For the evaluation of radiation exposure, dose-length-product (DLP) values were compared and Size-specific dose estimates (SSDE) values were calculated.
Diagnostic image quality was obtained from all patients. The median DLP (703.5 mGy · cm, range 390–2203 mGy · cm) was 7.9% lower when using the algorithm compared with the standard 120 kV protocol (median 756 mGy · cm, range 345–2267 mGy · cm). A reduction in potential to 100 kV occurred in 32 cases; therefore, these patients received significantly lower radiation exposure compared with the 120 kV protocol.
Automated attenuation-based tube potential selection produces good diagnostic image quality in chest-abdomen-pelvis CT and reduces the patient’s overall radiation dose by 7.9% compared to the standard 120 kV protocol.
KeywordsMultidetector computed tomography Cancer Cancer staging Neoplasms
In the clinic, the number of computed tomography (CT) examinations is increasing steadily ,. Compared with most other imaging modalities, CT imaging involves the use of increased radiation exposure . In CT examinations as well as in all other examination modalities, the ‘as low as reasonably achievable’ (ALARA) rule has to be considered, especially when radiation is applied to the patient. However, not only are the number of CT examinations increasing, the examination volume (e.g. chest-abdomen-pelvis examination) is increasing, too . In terms of staging cancer, the patient’s cumulative radiation exposure might cause problems in the future . In some cases, cancer is already the limiting disease, and therefore the benefits of staging cancer using radiation exposure outweigh the risks . However, some cancer patients—especially those in the early stages—will live long enough that the long-term effects of radiation become significant . Many techniques are already in use to minimise the radiation exposure of CT examinations: Automated attenuation-based tube current and voltage modulation as well as noise reduction filters and iterative reconstruction algorithms are current options ,.
Since dose modulation software for the adjustment of tube current was introduced, it has been used routinely all over the world and remains an important invention for reducing the radiation dose from imaging techniques ,–. Automated X-ray tube potential selection is also providing clinical radiologists and technicians another opportunity to adapt the radiation dose of the CT examination to the requirements of the specific body region –.
The rationale behind automated tube potential selection is that the contrast can be improved using lower X-ray tube potential, because low-energy X-rays are better absorbed than high-energy X-rays . For larger patients, the tube potential sometimes has to be adjusted to higher levels due to increased absorption occurring at a low tube potential .
Since automated attenuation-based tube potential selection was introduced in 2011, the technique has helped to lower radiation exposure, presumably as well as the automated tube current modulation has done, and continues to do so . A limited number of studies have investigated radiation exposure while using automated tube potential selection –,. One of the first trials reported using automated tube potential selection was performed for the imaging of the great vessels; in this case, it was possible to lower radiation exposure by 25.1% . This may be too much of a reduction for chest-abdomen-pelvic CT examinations, so the aim of our study was to evaluate automated tube potential selection for chest-abdomen-pelvic CT examinations carried out for staging reasons in cancer patients.
The study was performed as a single-centre, observer-blind study. The Institutional Review Board of our University Clinic (Goethe University Clinic, Frankfurt, Germany) approved this study; written informed consent requirement was waived since CAREkV is routinely used in all patients undergoing clinically indicated CT in our department. The data from consecutive unselected patients who underwent clinically indicated chest-abdomen-pelvis staging CT between January 2011 and March 2012 were analysed. The general exclusion criteria for contrast-enhanced CT included impaired renal function (estimated glomerular filtration rate <60 mL/min, calculated by creatinine blood level and patient age), hyperthyroidism, as well as hypersensitivity to iodine contrast media.
Study population and examination parameters
Slice · collimation (mm)
128 · 0.6
128 · 0.6
Patients 100 kV
Patients 120 kV
Patients 140 kV
Study population – indications and pathologies
Non-small-cell lung carcinoma (NSCLC)
Renal cell carcinoma
Urothelial cell carcinoma
Small cell lung cancer (SCLC)
Merkel cell carcinoma
Automated attenuation-based kV selection
The automatically selected kV remained stable throughout the whole examination. After that, the corresponding current, in mAs, was calculated. This was selected in order to attain the user-given image quality (e.g., 120 kV, 210 Ref.mAs). The user of the instrument provided the machine with a preferred image quality, which mostly depended on the standards in the particular radiology department. Then, the machine automatically selected the voltage and current, according to the type of examination being performed (e.g., vascular, bone, parenchymal, Figure 1). In our study, we used the algorithm for parenchymal organs because the CT examinations were performed for cancer staging reasons (Figure 1).
CT examinations were performed on the experimental group using a 128-slice CT machine (SOMATOM Definition Flash, Siemens, Forchheim, Germany), using automated attenuation-based tube potential selection (CAREkV). In the baseline group (conventional group) without CAREkV, only CareDose4D and a fixed tube potential of 120 kV were used.
Before the examination, the patients were enterally administered contrast media (Micropaque, Guerbet, Villepinte, France) and during the examination an intravenous contrast material, containing 1 ml/kg of iodinated contrast material (Ultravist 370, Bayer-Schering, Germany), followed by a saline chaser of 40 mL.
Contrast material, as well as saline flush, was injected at 2 mL/s into an antecubital vein using a double-syringe power injector (CT2, Medtron, Saarbruecken, Germany). A fixed delay of 70 seconds post injection was used in order to obtain venous contrast. All CT examinations were performed in a cranio-caudal direction starting from the upper thorax aperture down to the femoral ligaments, at a collimation of 128 · 0.6 mm, pitch 1.2, and rotation time of 0.5 seconds.
CT examinations were performed as follows:
Group 1 (experimental group) – 120 kV (using CAREkV – where kV can be automatically adapted to any of the following values: 80, 100, 120, and 140 kV), 210 Ref.mAs.
Group 2 (conventional group) – 120 kV (fixed), 210 Ref.mAs (using CareDose4D);
CT data reconstruction
For fast overviewing, images were reconstructed in 5-mm slice thickness using a 5-mm increment. For further evaluation, data were reconstructed with a slice thickness of 2 mm and increment of 1.5 mm using a medium-smooth soft-tissue convolution kernel (B30f) for parenchymal analysis. A hard convolution kernel was used for the analysis of bones and lungs (B70f). For detailed analysis and post-processing, images were transferred to an external workstation (SyngoVia, Multi-Modality Workplace, Siemens Healthcare, Forchheim, Germany).
Radiation dose estimations
Overview of patient characteristics
Group 1 vs. Group 2
Body diameter: transverse (cm)
Body diameter: lateral (cm)
Scanning range (cm)
CTDIvol (mGy · cm)
26.4 (3.9 – 58.9)
32.2 (19 – 52.3)
17.7 (3.7 – 42.5)
19.8 (12.1 – 34.1)
13.3 (9.7 – 28.6)
14.8 (9.4 – 28.9)
Detailed overview of the different examination protocols
Overall- Eff. mAs
Group 1 vs. Group 2
Group 1 vs. Group 2
Group 1 vs. Group 2
Patients 100 kV
p < 0.001
p = 0.6
p < 0.001
Patients 120 kV
Patients 140 kV
Patients 120 kV
Subjective image assessment was graded by two radiologists in consensus (with 2 and 5 years of experience in whole body imaging), applying a five-point scoring system: 1 = excellent: excellent definition of tumour and/or metastases, excellent delineation of the structures; 2 = good: good definition of tumour and/or metastases, minimal image noise; 3 = adequate: adequate definition of tumour and/or metastases, slight impact of image noise, sufficient for diagnosis; 4 = poor: poor definition of tumour and/or metastases, low attenuation and difficult delineation of the structures, increased image noise, diagnostic confidence reduced; 5 = unacceptable/nondiagnostic. The most probable reasons for reduced image quality were noted. Factors reducing image quality (obesity, motion, metallic artefacts, contrast medium flow-related, and contrast timing) were recorded by the radiologists.
Objective image quality (e.g., attenuation, noise, CNR) analysis was performed by one radiologist with 5 years of experience in general radiology on a regular PACS workstation (Centricity 4.2, General Electric Healthcare, Munich, Germany). The measurements were performed on several anatomic regions of the body (aorta at the level of the pulmonary trunk, both lobes of the lung, left lobe of the liver, right lobe of the liver, pancreas, spleen, kidneys, gluteus maximus muscle, bone and pre-sternally in the air). For calculation of SNR, image noise (or background noise) was determined as the standard deviation (SD) of air measured pre-sternally in front of the patient at the level of the ascending aorta. Based on these measurements, the signal-to-noise ratio (SNR) was determined according to the following equation: SNR = attenuation/background noise (Table 3). The CNR was defined as the difference in signal intensity between the venous attenuation in the vascular system (aorta) and the gluteus maximus muscle divided by the background noise. CNR was calculated as follows: CNR = [(ROIaorta – ROImuscle)/background noise]. To minimise bias from single measurements, we calculated the average of four measurements for each ROI.
Correlation analysis of DLP (mGy-cm) and patient diameter
DLP (mGy · cm)
Body diameter (mm): lateral
Body diameter (mm): transverse
Spearman correlation coefficient (rho) – lateral diameter/DLP
Spearman correlation coefficient (rho) – transverse diameter/DLP
Patients 100 kV
Patients 120 kV
Patients 140 kV
Patients 120 kV
Statistical analysis was performed using dedicated software (Bias for Windows 9.14; Epsilon, Germany). Continuous variables were reported as median and range, categorical variables as frequencies or percentages.
Radiation parameters and quantitative image parameters (e.g. noise, attenuation) were tested using the Wilcoxon Mann-Whitney U test as the data were nonparametric. The relationship between patient diameter and automated kV selection was analysed using the Spearman rank order correlation test. The Chi-square (X2) test was used for categorical variables (demographic patient data). Statistical significance was defined as a p-value above 0.05.
Radiation dose estimation
Automated attenuation-based tube potential selection resulted in a kV reduction of 120 to 100 kV in the experimental group, compared with the conventional group in 32/55 (58.2%) patients; it was kept stable at 120 kV in 17/55 (30.9%) patients and increased to 140 kV in 6/55 (10.9%) patients.
The difference of DLP between the groups (CAREkV vs. non-CAREkV) did not reach statistical significance (p = 0.43) (Figure 2).
Comparing our radiation exposure data from the experimental and conventional groups using SSDE calculations, it was clear that there was no significant difference (Table 3). However, the inter-group analysis showed that there was a significant difference between the 100 kV group compared to the 120 kV group (p < 0.001) and to the 140 kV group (p < 0.001) (Table 4). Between the 120 kV group and the 140 kV group there was no significant difference in SSDE values (p = 0.1).
However, this inter-group analysis should be examined with caution because there were only 6 patients in the 140 kV group. However, the results from the 100 kV group show that 32 patients were exposed to significantly less radiation, although the overall difference between the experimental and the conventional group did not reach statistical significance (SSDE p-value = 0.6; Table 3).
Patient diameter was similar in the CAREkV group and the conventional group, with a median lateral diameter of 34.4 cm (range 28.1–47.8 cm) and 33.1 cm (range 19.6–49.6 cm), respectively (Table 1).
Diagnostic image quality was obtained from all patients (excellent: n = 42; good: n = 9; moderate: n = 4). The reasons for moderate image quality (120 kV, n = 1; 100 kV, n = 3) were, in all cases, due to difficulties in ruling out parenchymal lesions, some because of image noise, and some because of insufficiencies concerning the contrast.
Concerning image quality in the control group, it was also rated as sufficient in all cases (excellent: n = 45; good: n = 7; moderate: n = 3).
Since the introduction of automated attenuation-based tube potential selection, most of the literature about it has been in the context of vascular imaging ,. In fact, only two studies have been published with a focus on cancer patients ,.
Of course, this is not the only technique available for lowering radiation exposure, and many new techniques are in use today; automated tube-currents as well as automated tube- potential selection are only two of many components that can be modified for reducing radiation exposure. Other interesting techniques like iterative reconstruction are also in use and have been shown to have additional potential in reducing radiation exposure . We wanted to evaluate whether this new technique would be valuable for the imaging of cancer patients concerning a non-stop (one-stop-shop) chest-abdominal-pelvis CT examination in a daily clinical routine. We evaluated a number of patients undergoing chest-abdomen-pelvis CT for cancer staging.
In CT, a reduction in radiation dose by lowering the tube potential results in a higher attenuation of the iodinated contrast media. However, this does not always have a positive effect, as the image noise will increase and soft-tissue contrast might be lost. To avoid this, the tube current has to be adapted to the lower kV setting (often to higher values). Therefore, the image noise will be comparable, but the radiation exposure will be lower compared to a 120 or 140 kV setting (Tables 3 and 4, Figure 2).
Winklehner et al.  evaluated automated attenuation-based kV selection in 40 patients for CT angiography of the aorta. In this study, an overall radiation dose reduction of 25.1% was observed, while keeping the image quality stable, when using a 120 kV protocol.
Eller et al.  evaluated automated attenuation-based kV selection in 100 patients. They carried out an abdominal CT examination for 52 of the patients, and a thoraco-abdominal examination for 48 patients. All examinations using automated attenuation-based kV selection resulted in a radiation dose reduction of at least 11.4%; in detail, 13.2% in the abdominal CT group and 9.5% in the thoraco-abdominal group.
Gnannt et al.  assessed automatic attenuation-based kV selection in 40 patients suffering from testicular cancer. In this study, a CT scan of the chest was performed in a mixed arterio-venous phase and the abdominal CT examination was carried out in the portal-venous phase of enhancement. The overall dose reduction was 12% on average. In all the studies mentioned above, there was no statistically significant worsening of subjective image quality.
In the study by Eller et al. , there was a dose reduction of 9.5% in thoraco-abdominal CT examination. Our data showed an overall dose reduction in 7.9% of the study population; this might be caused by different patient characteristics (e.g., body size and habitus). The tube potential in our study switched more often to 140 kV compared with the cited studies, and a tube potential of 80 kV was not automatically chosen in any case (Tables 1 and 4).
In the imaging of vessels, it is possible to examine the region of interest using a lower kV setting because of the high-contrast situation attained by the arterial phase of the contrast material. In clinical settings where parenchymal contrast is the object in question, for example, when parenchymal liver lesions have to be ruled out, automated attenuation-based tube potential selection might not lower the kV in the same aggressive manner as in the imaging of vessels (Figure 1).
Applying the algorithm to our study population, the overall radiation dose reduction was 7.9%. Comparing the DLP and SSDE values in the experimental group, it can be seen that the 100 kV patients received a significantly lower dose compared with both other groups (31 patients received an examination at the 100 kV setting, 17 at 120 kV, and only 7 at 140 kV).
In summary, these data indicate that in 31 patients, radiation exposure was significantly reduced, whereas in 24 (17 + 7) patients, radiation exposure remained the same or was not significantly increased. Keeping this in mind, more than half of the examined patients could potentially be examined with a lower radiation exposure in daily clinical routine. In a busy radiology department, sometimes it is difficult to adapt a CT examination protocol in detail to each patient, therefore, software such as CAREkV is able to improve the workflow and reduce radiation exposure.
Our study has some limitations. First, the overall number of patients was limited and further studies using a larger population are required. Second, we did not record the body mass index in detail, but added the patient diameter to this study as an alternative. Third, CAREkV is able to reduce the tube potential to as low as 80 kV. This tube potential, however, was not selected by CAREkV in our patient cohort. One cause for this might be the adult patient cohort, with a relatively large patient diameter on average. Also, good parenchymal contrast is needed in CT imaging for cancer staging.
In summary, we showed that attenuation-based tube potential selection is a good tool for dose reduction in daily clinical routine when ‘one-stop-shop’ cancer staging is performed. It leads to a good and stable overall image quality with only a mild increase in image noise. Radiation exposure was decreased by 7.9% compared to our former standard protocol using 120 kV.
Automated attenuation-based tube potential selection is easy to use and contributes to dose reduction efforts in daily clinical routine.
When comparing 120 kV fixed tube potential using automated tube current modulation and automated attenuation-based tube potential selection, both lead to the same image quality.
- Albert JM: Radiation risk from CT: implications for cancer screening. AJR Am J Roentgenol 2013, 201: W81–87. 10.2214/AJR.12.9226PubMedView ArticleGoogle Scholar
- Shah KH, Slovis BH, Runde D, Godbout B, Newman DH, Lee J: Radiation exposure among patients with the highest CT scan utilization in the emergency department. Emergency radiology 2013, 20: 485–491. 10.1007/s10140-013-1142-8PubMedView ArticleGoogle Scholar
- Brenner DJ, Hall EJ: Computed tomography–an increasing source of radiation exposure. N Engl J Med 2007, 357: 2277–2284. 10.1056/NEJMra072149PubMedView ArticleGoogle Scholar
- Saika K, Machii R: Five-year relative survival rate of cancer in the USA, Europe and Japan. Jpn J Clin Oncol 2013, 43: 1053–1054. 10.1093/jjco/hyt153PubMedView ArticleGoogle Scholar
- Lozano R, Naghavi M, Foreman K, Lim S, Shibuya K, Aboyans V, Abraham J, Adair T, Aggarwal R, Ahn SY, Alvarado M, Anderson HR, Anderson LM, Andrews KG, Atkinson C, Baddour LM, Barker-Collo S, Bartels DH, Bell ML, Benjamin EJ, Bennett D, Bhalla K, Bikbov B, Bin Abdulhak A, Birbeck G, Blyth F, Bolliger I, Boufous S, Bucello C, Burch M, et al.: Global and regional mortality from 235 causes of death for 20 age groups in 1990 and 2010: a systematic analysis for the Global Burden of Disease Study 2010. Lancet 2012, 380: 2095–2128. 10.1016/S0140-6736(12)61728-0PubMedView ArticleGoogle Scholar
- Fleischmann D, Boas FE: Computed tomography–old ideas and new technology. Eur Radiol 2011, 21: 510–517. 10.1007/s00330-011-2056-zPubMedView ArticleGoogle Scholar
- Greess H, Nomayr A, Wolf H, Baum U, Lell M, Bowing B, Kalender W, Bautz WA: Dose reduction in CT examination of children by an attenuation-based on-line modulation of tube current (CARE Dose). Eur Radiol 2002, 12: 1571–1576. 10.1007/s00330-001-1255-4PubMedView ArticleGoogle Scholar
- Gies M, Kalender WA, Wolf H, Suess C: Dose reduction in CT by anatomically adapted tube current modulation. I. Simulation studies. Med Phys 1999, 26: 2235–2247. 10.1118/1.598779PubMedView ArticleGoogle Scholar
- Kalender WA, Wolf H, Suess C: Dose reduction in CT by anatomically adapted tube current modulation. II. Phantom measurements. Med Phys 1999, 26: 2248–2253. 10.1118/1.598738PubMedView ArticleGoogle Scholar
- Kalra MK, Maher MM, Toth TL, Schmidt B, Westerman BL, Morgan HT, Saini S: Techniques and applications of automatic tube current modulation for CT. Radiology 2004, 233: 649–657. 10.1148/radiol.2333031150PubMedView ArticleGoogle Scholar
- McCollough CH, Bruesewitz MR, Kofler JM Jr: CT dose reduction and dose management tools: overview of available options. Radiographics 2006, 26: 503–512. 10.1148/rg.262055138PubMedView ArticleGoogle Scholar
- Siegel MJ, Ramirez-Giraldo JC, Hildebolt C, Bradley D, Schmidt B: Automated low-kilovoltage selection in pediatric computed tomography angiography: phantom study evaluating effects on radiation dose and image quality. Invest Radiol 2013, 48: 584–589. 10.1097/RLI.0b013e318289f918PubMedView ArticleGoogle Scholar
- Raman SP, Johnson PT, Deshmukh S, Mahesh M, Grant KL, Fishman EK: CT dose reduction applications: available tools on the latest generation of CT scanners. J Am Coll Radiol 2013, 10: 37–41. 10.1016/j.jacr.2012.06.025PubMedView ArticleGoogle Scholar
- Hough DM, Fletcher JG, Grant KL, Fidler JL, Yu L, Geske JR, Carter RE, Raupach R, Schmidt B, Flohr T, McCollough CH: Lowering kilovoltage to reduce radiation dose in contrast-enhanced abdominal CT: initial assessment of a prototype automated kilovoltage selection tool. AJR Am J Roentgenol 2012, 199: 1070–1077. 10.2214/AJR.12.8637PubMedView ArticleGoogle Scholar
- Eller A, May MS, Scharf M, Schmid A, Kuefner M, Uder M, Lell MM: Attenuation-based automatic kilovolt selection in abdominal computed tomography: effects on radiation exposure and image quality. Invest Radiol 2012, 47: 559–565. 10.1097/RLI.0b013e318260c5d6PubMedView ArticleGoogle Scholar
- Gnannt R, Winklehner A, Eberli D, Knuth A, Frauenfelder T, Alkadhi H: Automated tube potential selection for standard chest and abdominal CT in follow-up patients with testicular cancer: comparison with fixed tube potential. Eur Radiol 2012, 22: 1937–1945. 10.1007/s00330-012-2453-yPubMedView ArticleGoogle Scholar
- Niemann T, Henry S, Faivre JB, Yasunaga K, Bendaoud S, Simeone A, Remy J, Duhamel A, Flohr T, Remy-Jardin M: Clinical evaluation of automatic tube voltage selection in chest CT angiography. Eur Radiol 2013, 23: 2643–2651. 10.1007/s00330-013-2887-xPubMedView ArticleGoogle Scholar
- Winklehner A, Goetti R, Baumueller S, Karlo C, Schmidt B, Raupach R, Flohr T, Frauenfelder T, Alkadhi H: Automated attenuation-based tube potential selection for thoracoabdominal computed tomography angiography: improved dose effectiveness. Invest Radiol 2011, 46: 767–773. 10.1097/RLI.0b013e3182266448PubMedView ArticleGoogle Scholar
- AAPM Task Group: Size-Specific Dose Estimates (SSDE) in pediatric and adult body CT examinations. American Association of Pysicists in Medicine 2011, 1–22.Google Scholar
- Christner JA, Braun NN, Jacobsen MC, Carter RE, Kofler JM, McCollough CH: Size-specific dose estimates for adult patients at CT of the Torso. Radiology 2012, 265: 841–847. 10.1148/radiol.12112365PubMedView ArticleGoogle Scholar
- Eisenhauer EA, Therasse P, Bogaerts J, Schwartz LH, Sargent D, Ford R, Dancey J, Arbuck S, Gwyther S, Mooney M, Rubinstein L, Shankar L, Dodd L, Kaplan R, Lacombe D, Verweij J: New response evaluation criteria in solid tumours: revised RECIST guideline (version 1.1). Eur J Cancer 2009, 45: 228–247. 10.1016/j.ejca.2008.10.026PubMedView ArticleGoogle Scholar
- Goetti R, Winklehner A, Gordic S, Baumueller S, Karlo CA, Frauenfelder T, Alkadhi H: Automated attenuation-based kilovoltage selection: preliminary observations in patients after endovascular aneurysm repair of the abdominal aorta. AJR Am J Roentgenol 2012, 199: W380–385. 10.2214/AJR.11.8029PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.